1. Easy to assemble and calibrate, high reliability
2. Fully encapsulated, water proof moisture proof and lightning protection
3. Surge protection
4. Operating temperature: -20°C to +85°C
Housing material: stainless steel
Type: 4-wire system
Balancing adjustment steps
1. Connect the load cell wire to the junction box according to permutation, obtain the corresponding relation with the junction box terminal and adjustable resistance. When in need of adjusting angular difference, rotate the corresponding potentiometer to eliminate the angular difference.
2. Put a weight on the four angles in sequence, record the weight value of the four load cells display in the indicator, and order lowest to sort.
3. Screw the potentiometer in counterclockwise direction of the load cell wire with largest value, and screw the potentiometer in clockwise direction of the load cell wire with smallest value.(If only exist one largest/smallest value, and the other three are the same, just adjust one corresponding potentiometer.
4. Repeat step 2 and step 3 until the four weight value are the same.